|Minimum Order Quantity||1 Unit|
|Airflow||3 lpm & 30 Lpm|
|Operation Time||1 HOURS|
|Accuracy||+/- 5 %|
|Air Flow Rate||AS PER REQ|
Stack Temperature Range : Ambient to 600 deg C read on a Digital Pyrometer.
Stack Velocity Range : 3 to 60 m/sec
Thermocouple: T/C sensor in SS 304 casing, length of insertion: 1 m with 2m long cable.
Manometer : Digital with 0-1300 mm of H2O range
Pitot Tube : Calibrated S-type fabricated -SS 304, 1m long
Particulate Sampling : 0 -60 lpm collection on thimble type filter up to 0.3 micron rating.
Gaseous Sampling : 0 - 3 lpm collection in a set of Borosilicate glass impingers.
Rotameter : Acrylic body with 5% FSD accuracy, 0 - 30 lpm for PM and 0-3 lpm for GAS.
Sampling Probe : 1m long SS tube with inside lining of Teflon & provision for controlled at 120 deg C heating.
Filter Holder : Glass filter holder with heating provision suitable to hold either cellulose filteration
thimble (size 28mm lD X 100mm long) or glass fibre thimble (size 19mm lD X 90mm long). Nozzles : A set of 3 glass nozzles.
Digital Clock : 0-60 minutes, 1 second readout with start and stop switches.
Temp. Controller : Digital type temperature controller.
Dry Gas Meter : Range 0 to 40 LPM connected to SPM line.
Sampling Train : 3 Nos. of 240 ml cap glass impinger, 1 No. condenser glass coil, 1 No. moisture bottle 400 ml & 1 No. cartridge tube accommodated in separate ice tray with instrument panel with a provision to cool down flue gas temperature.
Vacuum Pump : Monoblock Rotary Vane type, oil free, 0.5 HP single phase motor (230V) with more than 100 lpm free flow capacity.
List of Components GTI-651:
(1) :Instrument panel consisting of pyrometer, manometer rotameters, vacuum gauge, stop watch, dry gas meter & temperature controller.
(2): Probe kit consisting of probe pipe, pitot tube, connecting hoses, tool kit, filter holder, nozzles, thermocouple, heating system, tripod etc.
(3): Cold box assembly consisting of glass impingers, condenser glass coil, moisture bottle & water pump.
(4): Vacuum pump assembly.
Reviewed by 6 Users
Most Relevant Reviews